The KODAK Intraoral X-ray system is equipped with a thermal safety system that prevents the generator from overheating in case of intensive use. Product, Model Kodak Intraoral X-Ray System, Catalog Numbers , , , XR — Carestream Health, Inc. Made in France Kodak Intraoral X-Ray System Performance. Operational ease. Safety. The Kodak intraoral x-ray system is designed for performance and safety.
|Published (Last):||4 April 2007|
|PDF File Size:||9.3 Mb|
|ePub File Size:||16.39 Mb|
|Price:||Free* [*Free Regsitration Required]|
Auth with social syxtem I would recommend you not dismantle the ceramic resistor. It shows no numbers just KV Error. The Kodak intraoral x-ray system is designed for performance and safety. The new wall mounting unit enables IRIX customers to upgrade without completely dismantling their current setup, providing them with all the advantages of the new system including higher image quality, reduced dose and more options for better diagnosis.
For next procedure for self-calibration: In this case there’s no high voltage and you have KV error. Hello My Kodak is always giving the message of KV error. Unless Trophy designed it with certain service syystem of each unit. Once the old solder removed, the wires soldered back with new solder do not overheat place the high voltage contact spring and insert the high voltage multiplier block inside the plastic cup containing the feedback resistor.
I opened the tube collect all that oil. The future of periodontal therapy.
Kodak Kv/error – Service Technicians Forum
Sign In or Register. And, after each exposure, the dose level is displayed for simple dose monitoring. Content you may be interested in: Hello, Concerning the HV feddback resistor, I have 3 ceramic high voltage feedback resistors in case of emergency assistance, also have the high voltage block with the xray tube and multiplier, please contact for assistance details.
Hi, In case you opened the tubehead to change the ceramic resistance that burned due to high voltage spark, the rectifier diodes or capacitors for the multiplier might also be damaged. December 3, Live Webinar: The Online Medical Device Exhibition.
Kodak Intraoral X-Ray System | Pre-Owned Dental Inc.
It also offers a new mounting configuration for users of the popular Trophy IRIX X-ray systems — providing a convenient, cost-effective upgrade option for these customers. Log in or Register to rate and post a comment. It gives KV Error along with bip and on the control panel two green lights turn on. The resistor is very fragile!
I did the own PBC board that had pcs of 1Mohm smd resistors in the form of x-rqy snake and 13 pcd of 1kOhm smd resistors were connected in series. Changes in ABO certification: Please, I need the service manual.
Hi Faustino, Do you syystem ceramic resistor 13 Kohm? The voltage devider must be immersed in oil completely, no air bubbles present in oil! Complete control The Kodak system is equipped with adjustable tension 60 or 70 kV to accentuate contrast and levels of gray okdak according to your diagnostic needs. Thanks, I will forward your email for those who need help on tube repairs and software.
Could you please send it to msobota gmx. As a part of its mission to always provide dentists with only the best solutions for improved patient 220, ContacEZ continues to refine, improve and expand its product lines to make them safer and more effective than effort.
Interested in the regional page in your own language?
Used & Refurbished Dental Equipment
My panel is without rotating timer knob. December 21, ADA taps Dr. Other information will follow by mail.